Laboratory Equipment Desktop Sterile Hood Small Cabinet
|FOB Price:||US $180-330 / Piece|
|Min. Order:||1 Piece|
|Min. Order||FOB Price|
|1 Piece||US $180-330/ Piece|
|Production Capacity:||1000 Set / Month|
|Payment Terms:||T/T, Western Union|
- Model NO.: HD-650
- Type: Central Bench
- Color: Grey
- Condition: New
- Brand: Sujing
- Clean Grade: Grade 100(209e U.S. Federal)
- Working Area: 500*400*450mm
- Certificate: ISO9001
- Equiblast Speed: 0.25-0.45m/S
- Transport Package: Wooden
- Origin: China, Zhejiang
- Material: Stainless Steel
- Usage: Hospital, School, Company, Research Institute
- Customized: Customized
- Original Place: Zhejiang, China
- Air Supply: Horizontal Air Supply
- Overall Dimension(L*W*H): 650*550*850mm
- Power Supply: AC,220V/50Hz
- Noise: 62dB
- Trademark: Sujing
- Specification: GMP
- HS Code: 9031200090
It is widely used in medicine, pharmacy, biological & chemical experiment, electron, precise apparatus, instrument and so on.
1. Level air flow clean bench
2. It is widely used in medicine, pharmacy, biological & chemical experiment, electron, precise apparatus, instrument and so on
3. Contact switch adjusts voltage to keep wind speed in working area in ideal state all the time.
Note: Special specification of non-standard type clean bench is also available on demand.
|Clean grade||Grade 100(209E U.S. Federal)|
|Number of bacteria||<0.5per utensil.hour(90mm utensil)|
|Dimension of working area(W1×D1×H1)||500*400*450mm|
|Specification and number of light|
from firefly/ ultraviolet light
|Specification and number of|
high effective filter
1,based on actual usage, periodically remove the primary filter cleaning, the cleaning cycle is generally 3-6 months. (if not long-term cleaning, the dust will affect the amount of wind insufficient to reduce the cleaning effect.)
2, when the early effect of the normal exchange or clean the filter air filter, still can not achieve the desired cross-section wind speed, then adjust the fan voltage, and thus achieve the desired uniform wind speed.
3,generally when the fan operating voltage adjustment to the highest point in the use of eighteen, still can not achieve the desired wind speed, description of the filter on the high efficiency filter dust too much (filter kong has basically stuck to timely update) general efficiency air filters use a period of eighteen months.
4,replacement of high efficiency air filters, should pay attention to the correctness of the model dimensions (original manufacturer configuration), arrow wind devices, and pay attention to the peripheral sealing of the filter is absolutely no leakage occurs.